|Supported pipe materials||Metal pipes, Plastic pipes (soft/hard) *1|
|Supported fluids||Liquids (water, oil, adhesive, grease, chemical solutions, etc.) *1|
|Supported fluid temperature||0 °C (no freezing on the pipe surface) to 100 °C 32 to 212 °F (Pipe surface temperature)|
|Clamp set model||Plastic pipe/tube attachment||FD-XC1R1, FD-XC1R2|
|Metal pipe attachment||FD-XC1M*2|
|Maximum rated flow rate||0 to 1000 mL/min|
|Zero cut flow rate||20 mL/min*3 (variable, default)|
|Display resolution||Instantaneous flow rate||0.1/1/10 mL/min (Displayed on controller)|
|Shot amount||0.001/0.01/0.1/1 mL (Displayed on controller)|
|Repeatability||Plastic pipe/tube attachment||Response time: 50 ms||F.S.||± 0.6 %*4*5|
|Instantaneous flow rate||±6 mL/min*4*5|
|Response time: 500 ms||±1.9 mL/min*4|
|Metal pipe attachment||Response time: 50 ms||F.S.||± 1 %*4*5|
|Instantaneous flow rate||±10 mL/min*4*5|
|Response time: 500 ms||±3.2 mL/min*4|
|Integrated unit display||0.1/1/10/100/1000/10000 mL (displayed on controller)|
|Display method||Status indicator|
|Environmental resistance||Enclosure rating||IP65/IP67 (IEC60529) , IP68G (JIS C0920) *6|
|Ambient temperature||0 to 60 °C (No freezing) 32 to 140 °F|
|Relative humidity||35 % to 85 %RH (No condensation)|
|Vibration resistance||10 to 55 Hz, double amplitude 1.5 mm 0.06 “, 2 hours each for X,Y,Z direction|
|Shock resistance||50 G 11 ms 3 times each for X,Y,Z direction|
|Material||Sensor head||Head body: PPS/PPSU, in-cable amplifier: PPS, cable: PVC, controller connector: PPS/PBT/POM|
|Clamp set||For plastic pipe||Body, fixing screw: PPS, detection surface: special rubber, pipe support rubber: FKM, sensor head fixing screw: SUSXM7|
|For metal pipe||Metal: SUS304/SUSXM7, detection surface: special rubber, clamp support rubber: FKM, sensor head fixing screw: SUSXM7|
|Weight||Approx. 230 g|
|*1 Liquid must allow for the passage of an ultrasonic pulse, as well as not contain large air pockets or excessive bubbles. Readings may become unstable depending on the type of pipe.
*2 When using stainless steel or iron pipes, the ideal pipe wall thickness is as follows, FD-XS1: approx. 0.5 mm 0.02″, FD-XS8: approx. 1 mm 0.04″, FD-XS20: approx. 1 0.04″ to 2 mm 0.08″. FD-X signal strength and stability will decrease as the thickness of the pipe wall increases or decreases from the suggested size.
*3 The zero cut flow rate can be changed in the settings. When using the unit with a low flow rate range, perform an origin adjustment when the fluid is not moving if you change the zero cut flow rate.
*4 This specification is valid when the flow velocity distribution is stable. This value does not take into account the effects of pulsation or fluctuations in flow velocity distribution due to facility factors. Convert the F.S. (full scale value) listed in the table according to the rated flow range.
*5 The longer the response time is set, the more repeatability is improved. As a guideline, use √ (50 ms/response time) times.
*6 The connector part of the sensor head cable is IP65/IP67.
- Thương hiệu
- Tự động hóa
- Đo lường & điều khiển
- Thiết bị đóng cắt
- Vật tư tủ điện
- Khí nén
- Điện dân dụng
- Dịch vụ